Volume 10, Number 7—July 2004
SARS Coronavirus Detection
We developed a set of three real-time reverse transcription–polymerase chain reaction (PCR) assays that amplify three different regions of the SARS-associated coronavirus (SARS-CoV), can be run in parallel or in a single tube, and can detect <10 genome equivalents of SARS-CoV. The assays consider all currently available SARS-CoV sequences and are optimized for two prominent real-time PCR platforms.
Recently, a new coronavirus was identified as the suspected causative agent of an increased number of atypical pneumonia cases reported from Hong Kong, Singapore, Vietnam, and Canada (1–4). Subsequent publications demonstrated that this new coronavirus was detectable in patients with severe acute respiratory syndrome (SARS) (5,6), classified according to the World Health Organization’s case definition (7). During the first 6 months of 2003, a total of 8,422 patients were affected. This fact, together with the reappearance of the SARS-associated coronavirus (SARS-CoV) in China in late 2003, makes it clear that rapid and reliable diagnostic tools are essential for accurate disease reporting and subsequent disease management.
Because a defined treatment program and vaccination strategy are lacking, the main strategy to counteract the spread of this emerging virus is timely identification and isolation of infected persons. SARS patients’ typical initial symptoms include fever, cough, and headache, similar to many acute viral respiratory infections. Therefore, molecular-based diagnostic methods are applied to rapidly identify SARS-CoV–infected persons. Recently, nested and real-time reverse transcription–polymerase chain reaction (RT-PCR) assays to detect SARS-CoV have been published (5,8). These assays—the first tools to detect SARS-CoV in patients with SARS—were based on the short stretches of viral sequence identified as the RNA-directed RNA polymerase of a new microbe.
Subsequently, sequences from several SARS-CoV isolates were determined, and all of these sequences were closely related, as would be expected during the clustered outbreaks in 2003. However, the genomes of RNA viruses, including those of coronaviruses, tend to vary over time and with location (9–12). Recently, the sequence variations of SARS-CoV during the first epidemic phases in China in 2003 were reported. The neutral mutation rate for SARS-CoV was almost constant and similar to that of known RNA viruses; the S protein, responsible for virus-host receptor recognition, displayed the most extensive amino acid changes (13). In addition, the sequence analysis of isolates from recent SARS patients in China in 2004 has shown that 98.8%–99.4% of the 3,768 bases of S gene, 99% of 658 bases of M gene, and 99% of 1,068 bases of N gene are isogenous with those submitted to public databases, which date back to the first epidemic in spring 2003 (14). However, even these minimal changes could render existing PCR assays ineffective should SARS-CoV reemerge (15).
To improve the ability to detect SARS-CoV safely and reduce the risk of eliciting false-negative results caused by genome sequence variations, we established three individual real-time RT-PCR assays. Target sequences were chosen by using the following criteria: 1) the regions are distributed over the whole genome, including the nonstructural polyprotein 1a and 1ab genes and the spike glycoprotein gene (Table 1); 2) the regions are highly conserved among the 89, 90, and 100 respective sequences available in public sequence databases; 3) the regions are suitable for the design of a real-time RT-PCR assay; and 4) the designed primers, 5′-nuclease probes, and amplicons displayed no considerable homology to other viruses, including human CoV OC43 and 229E in BLAST searches (available from http://www.ncbi.nlm.nih.gov/BLAST/).
These assays were based on the fluorogenic oligoprobe chemistry, which uses the 5′-exonuclease activity of the DNA polymerase to generate a more specific signal than that produced by the use of SYBR Green I (8). The real-time RT-PCR assays were successfully run on the Applied Biosystems real-time PCR systems (SDS7700 and SDS7000; Applied Biosystems, Foster City, CA) as well as on the Roche LightCycler (Roche Diagnostics GmbH, Mannheim, Germany). All assays were designed as one-step RT-PCR reactions to be run under identical conditions on the respective PCR platform. This system allowed the simultaneous detection of different SARS-CoV regions in a single PCR run. Moreover, we could combine the three assays in a single tube, which might be important when clinical material is limited. Finally, the assays were compared to the 5′-nuclease assay published recently (5) and to a commercially available real-time PCR kit (Real-Art HPA-Coronavirus LC RT PCR Reagents, Artus GmbH, Hamburg, Germany).
After optimization of primer and 5′-nuclease probe concentration and annealing temperature, reaction conditions for our 5′-nuclease assay were as follows. For the RT-PCR performed on the Applied Biosystems platforms, each 25-μL reaction contained 12.5 μL of 2xQuantiTect Probe RT-PCR Master Mix (Qiagen, Hilden, Germany), 10 pmol of each primer, 3 pmol of 5′-nuclease probe, and 0.25 μL of QuantiTect Probe RT Mix. RNase-free water was added up to 23 μL, and 2 μL of RNA preparation was used. Cycling conditions were 30 min at 50°C for RT reaction, 15 min at 95°C for inactivation of RT, activation of the Taq DNA polymerase, and cDNA denaturation, followed by 45 cycles of 15 s at 95°C and 30 s at 60°C. Total running time was 140 min.
For LightCycler RT-PCR reactions, each 20-μL reaction included 7.5 μL of 2.7x LightCycler RNA Master Hybridization Probes mix (Roche Diagnostics GmbH), 10 pmol of each primer, 3 pmol of the 5′-nuclease probe, and 1.3 μL Mn (OAc)2 (50 mmol/L). RNase-free water was added up to 18 μL, and 2 μL of RNA was used. Cycling conditions were 20 min at 55°C for the RT reaction, 30 s at 95°C for initial denaturation, followed by 45 cycles of 1 s at 95°C, 10 s at 55°C, and 10 s at 72°C. Total running time was 55 min. The combined assays were set up by adding all primers and probes in the same concentration; the amount of water was reduced accordingly. Protocols are also available from the Robert Koch-Institut homepage (available from www.rki.de). The human L13 gene and the human cyclophilin gene (16) were amplified under identical reaction conditions as the SARS-CoV–specific assays on the ABI platforms and the LightCycler, respectively, to act as amplification controls.
To evaluate the sensitivity of the SARS-CoV–specific assays, RT-PCRs were performed repeatedly on serial dilutions of RNA extracted with the Viral RNA Kit (Qiagen) from cultured SARS-CoV with defined amounts of genome equivalents (GE) by using the international standard of the European Network for the Diagnostics of Imported Viral Diseases (ENIVD), distributed through the Robert Koch Institute (available from: http://www.rki.de/INFEKT/SARS/DATASHEET.PDF). Results are shown in Table 2. Detection limits of the three new assays were >10 GE. Comparison of the threshold cycle (CT) values showed that the new assays were at least as sensitive as the previously described assay (5,8) and the commercially available kit. When we combined the assays in a single tube targeting three different regions on the same RNA template, the CT was reduced by 1 to 2 cycles (SDS7700), which suggests either that sensitivity was unchanged or, when there was an increase, that it was attributable to the combination of all three signals (LightCycler). Subsequent agarose gel analysis during optimization steps of the PCR confirmed the simultaneous amplification of the three RT-PCR products (Figure, online only; available from: http://www.cdc.gov/ncid/EID/vol10no7/03-0678.htm).
Using the single or combined assays, we analyzed 27 bronchoalveolar-lavage fluid samples from 19 suspected SARS case-patients and 8 probable SARS patients (according to the Robert Koch-Institut case definition, available from http://www.rki.de/INFEKT/SARS/AOLG-FALLDEF-ARSUU.PDF). All samples were positive for L13 and cyclophilin control sequences when amplified in parallel. In agreement with the previously published assay results (5), SARS-CoV was detectable in three samples from eight probable SARS patients, without significant explicit differences in the CT value of individual samples when the single or combined assays were used. These patients were seropositive and are regarded as confirmed SARS patients. Respiratory samples and stool samples taken 8 days later from the remaining 5 probable patients as well as the 19 persons with suspected cases were negative by RT-PCR. Moreover, these patients remained seronegative and are regarded as unconfirmed SARS patients. In addition, 35 serum samples from patients with SARS-CoV infection obtained 1–52 days after disease onset were analyzed. Between 20 and 1,000 GE/mL of SARS-CoV specific RNA was detected in 21 of 35 serum samples, even when serum was obtained from patients 1 day after disease onset. (A detailed description of this study will be published later.)
None of the assays displayed cross-reactivity to clinical samples containing human cDNA from blood; human CoV 229E; influenza viruses A and B; parainfluenzaviruses 1,2, and 3; respiratory syncytial virus; rhinoviruses; enteroviruses; adenoviruses 1–10; human metapneumovirus; Mycoplasma pneumoniae; or Chlamydia pneumoniae. For these pathogens, we obtained neither a fluorescent signal nor an amplification product in subsequent agarose gel analysis (selection shown in online Figure). Although we focused on a one-step RT-PCR to decrease handling and total assay time, the three real-time RT-PCR assays can also be performed as two-step RT-PCR, including a separate cDNA synthesis step followed by PCR, and then finally using appropriate ready-to-use master mixes and the same cycling conditions and omitting the RT step.
The single assays and the combined assay were also used in an external quality assessment to detect SARS-COV, organized by the ENIVD. All assays could detect SARS-CoV in 7 of 11 samples with virus loads ranging from 5x106 to 2x103 GE of two isolates of SARS-CoV per milliliter sample without false-positive or false-negative results. While the application of three single assays to detect SARS-CoV leads to a higher reliability of negative results, reflecting the negative outcome of three independent amplification reactions, it is a more expensive approach than combining the assays.
In conclusion, the real-time RT-PCR assays we describe provide a fast and reliable tool that can complement and improve recently introduced techniques for SARS diagnostics. Parallel amplification of two human reference genes, L13 and cyclophilin, confirmed negative results in clinical samples by demonstrating amplifiable RNA. The separation of the control reaction was chosen to guarantee the high sensitivity of the SARS-CoV detection of <10 GE of SARS-CoV per reaction. An RT-PCR run is completed in <1 h, depending on the real-time PCR platform. In cases of small amounts of material or in an emergency situation with a high throughput of samples, the three SARS-CoV–specific assays can be combined into one RT-PCR reaction without loss of sensitivity. Furthermore, as the ambiguous diagnostic results in a hospital in Canada have recently shown (17), targeting three different regions distributed over the whole genome considerably reduces the risk for false-negative results caused by virus sequence modifications.
Dr. Nitsche is a research fellow at the Centre for Biological Safety of the Robert Koch-Institut, Berlin. His primary interest is the molecular detection of human pathogens with special focus on emerging viral infections.
We thank Sabrina Wendt for excellent technical assistance, Ian M. Mackay for critically reading the manuscript, and Artus (Artus GmbH, Hamburg, Germany) for kindly providing the Real-Art HPA-Coroniavirus LC RT PCR Reagents Kit.
- Peiris JS, Lai ST, Poon LL, Guan Y, Yam LY, Lim W, Coronavirus as a possible cause of severe acute respiratory syndrome. Lancet. 2003;361:1319–25.
- Poutanen SM, Low DE, Henry B, Finkelstein S, Rose D, Green K, Identification of severe acute respiratory syndrome in Canada. N Engl J Med. 2003;348:1995–2005.
- Tsang KW, Ho PL, Ooi GC, Yee WK, Wang T, Chan-Yeung M, A cluster of cases of severe acute respiratory syndrome in Hong Kong. N Engl J Med. 2003;348:1977–85.
- Acute respiratory syndrome. China, Hong Kong Special Administrative Region of China, and Viet Nam. Wkly Epidemiol Rec. 2003;78:73–4.
- Drosten C, Günther S, Preiser W, van der Werf S, Brodt HR, Becker S, Identification of a novel coronavirus in patients with severe acute respiratory syndrome. N Engl J Med. 2003;348:1967–76.
- Ksiazek TG, Erdman D, Goldsmith CS, Zaki SR, Peret; Emery S, et al. A novel coronavirus associated with severe acute respiratory syndrome. N Engl J Med. 2003;348:1953–66.
- World Health Organization. Case definition for surveillance of severe acute respiratory syndrome SARS. [cited 2003 May 1]. Available from: http://www.who.int/csr/sars/casedefinition/en/
- Poon LL, Wong OK, Chan KH, Luk W, Yuen KY, Peiris JS, Rapid diagnosis of a coronavirus associated with severe acute respiratory syndrome (SARS). Clin Chem. 2003;49:953–5.
- Pratelli A, Martella V, Elia G, Decaro N, Aliberti A, Buonavoglia D, Variation of the sequence in the gene encoding for transmembrane protein M of canine coronavirus (CCV). Mol Cell Probes. 2001;15:229–33.
- Vaughn EM, Halbur PG, Paul PS. Sequence comparison of porcine respiratory coronavirus isolates reveals heterogeneity in the S, 3, and 3-1 genes. J Virol. 1995;69:3176–84.
- Ruan YJ, Wie CL, Ee AL, Vega VB, Thoreau H, Su ST, Comparative full-length genome sequence analysis of 14 SARS coronavirus isolates and common mutations associated with putative origins of infection. Lancet. 2003;361:1779–85.
- Holmes KV, Lai MMC. Coronaviridae: The viruses and their replication. In: Fields BN, Knipe DM, Howley PRM, editors. Fields virology. 3rd ed. New York: Lippincott Williams & Wilkins; 2001. p. 1075–94.
- Chinese SARS Molecular Epidemiology Consortium. Molecular evolution of the SARS coronavirus during the course of the SARS epidemic in China. Science. 2004;303:1666–9. Epub 2004 Jan 29.
- People’s daily edited by ProMED-mail. SARS worldwide. ProMed January 2, 2004. Accessed at: http://www.promedmail.org, archive number: 20040102.0015. 19991013.1826.
- Wood L. questions about comparative genomics of SARS coronavirus isolates. Lancet. 2003;362:578–9.
- Radonic A, Thulke S, Mackay IM, Landt O, Siegert W, Nitsche A. A guideline to reference gene selection for quantitative real-time PCR. Biochem Biophys Res Commun. 2004;313:856–62.
- Kondro W. Canadian officials watch SARS-like mystery bug. Lancet. 2003;362:714.
Suggested citation for this article: Nitsche A, Schweiger B, Ellerbrok H, Niedrig M, Pauli G. SARS coronavirus detection. Emerg Infect Dis [serial on the Internet]. 2004 Jul [date cited]. Available from: http://wwwnc.cdc.gov/eid/article/10/7/03-0678.htm