Volume 25, Number 3—March 2019
Genomic Analysis of Cardiac Surgery–Associated Mycobacterium chimaera Infections, United States
Highlight and copy the desired format.
|EID||Hasan NA, Epperson L, Lawsin A, Rodger RR, Perkins KM, Halpin A, et al. Genomic Analysis of Cardiac Surgery–Associated Mycobacterium chimaera Infections, United States. Emerg Infect Dis. 2019;25(3):559-563. https://dx.doi.org/10.3201/eid2503.181282|
|AMA||Hasan NA, Epperson L, Lawsin A, et al. Genomic Analysis of Cardiac Surgery–Associated Mycobacterium chimaera Infections, United States. Emerging Infectious Diseases. 2019;25(3):559-563. doi:10.3201/eid2503.181282.|
|APA||Hasan, N. A., Epperson, L., Lawsin, A., Rodger, R. R., Perkins, K. M., Halpin, A....Strong, M. (2019). Genomic Analysis of Cardiac Surgery–Associated Mycobacterium chimaera Infections, United States. Emerging Infectious Diseases, 25(3), 559-563. https://dx.doi.org/10.3201/eid2503.181282.|
A surgical heater–cooler unit has been implicated as the source for Mycobacterium chimaera infections among cardiac surgery patients in several countries. We isolated M. chimaera from heater–cooler units and patient infections in the United States. Whole-genome sequencing corroborated a risk for these units acting as a reservoir for this pathogen.
Mycobacterium chimaera is a species in the Mycobacterium avium complex (MAC) (1). MAC is the most frequently reported cause of nontuberculous mycobacterium (NTM) infection in the United States, although disseminated M. chimaera infections are relatively rare (2). In 2012, investigators in Switzerland found that some patients with disseminated M. chimaera infections had undergone open-chest cardiac surgeries, during which they were exposed to heater–cooler units (HCUs) (3). These devices, Stöckert 3T Heater–Cooler Units (LivaNova PLC, https://www.livanova.com; formerly Sorin Group Deutschland GmbH), manufactured in Germany, were unknowingly contaminated with M. chimaera (4,5). In the same year, a Pennsylvania hospital identified a cluster of invasive M. chimaera infections among open-chest cardiac surgery patients exposed to LivaNova 3T HCUs contaminated with M. chimaera (6), which prompted notification of ≈1,300 patients with exposure to these units (7). Additional cases of disseminated M. chimaera infection among cardiac surgery patients have emerged worldwide, with evidence implicating bioaerosols produced by contaminated LivaNova 3T HCUs as the source of post–cardiac surgery M. chimaera infections (8,9). We report the relationships among HCU-associated isolates from patients and LivaNova 3T HCUs in the United States and their context among the global outbreak.
During 2015–2016, we collected NTM isolates from 3T HCU water (n = 38 isolates) and suspected patient cases (n = 24 isolates) from 8 US locations. We identified isolates and conducted high-throughput whole-genome sequencing using the Illumina Miseq system (https://www.illumina.com). We selected Pennsylvania isolate 2015-2271 (USA_PA_PAT_9) for Pacific Biosciences (https://www.pacb.com) single-molecule real-time sequencing (10). We downloaded publicly available M. chimaera genomes from isolates collected in Australia, Denmark, Italy, New Zealand, the United Kingdom, and Switzerland from the National Center for Biotechnology Information (NCBI) Sequence Read Archive (SRA). We included Zürich CHE_HCU_1 isolate as a representative of the genotype isolated from HCUs, patients, and manufacturing sites in Europe (9). For each isolate, we mapped the sequence reads to the M. chimaera strain CDC 2015-22-71 reference genome (GenBank accession no. NZ_CP019221.1) to detect single-nucleotide polymorphisms (SNPs) (Appendix).
We reconstructed phylogenetic relationships among M. chimaera isolates collected from post–cardiac surgery patients and HCUs in 8 locations across the United States, as well as HCU-associated strains from Australia, New Zealand, and Europe (Table; Appendix Figure 1). We compared all HCU-associated isolates with 7 M. chimaera respiratory isolates obtained from US patients with no history of cardiac surgery. We identified 18,190 SNPs in the 3.82-Mb core genome (62.8% of the reference genome) among 126 M. chimaera isolates.
The NeighborNet splitstree (Appendix) of M. chimaera showed 3 groups (HCU1, HCU2, and non-HCU; Figure 1). Clade HCU1 (n = 112 isolates; Figure 2) is a discrete cluster composed entirely of HCU-associated isolates from case-patients and HCUs (mean pairwise distance 4 SNPs, range 0–23 SNPs; Appendix Figure 2) from Australia, Denmark, New Zealand, Switzerland, the United Kingdom, and the United States. Clade HCU2 was composed of 3 HCU-associated M. chimaera isolates from Switzerland (2) and Australia (AUS_HCU_30 and AUS_HCU_31). The mean difference among HCU2 isolates was 21.3 SNPs (range 19–25 SNPs; Appendix). Clade 3 was composed of US non–HCU-associated isolates (non-HCU). Two HCU-associated patient isolates from Australia (mean non–HCU-associated isolate pairwise SNPs 52, range 3–111 SNPs; Appendix) were unclustered. The mean distance between HCU1 and international HCU isolates was 13.58 SNPs (range 0–521 SNPs; Appendix Figure 3); the mean distance between HCU1 and non–HCU-associated isolates was 510.5 SNPs (range 506–610 SNPs; Appendix). In comparison, the mean distance between HCU2 and non-HCU isolates was 17,130.7 SNPs (range 17,057–17,221 SNPs). Of the 117 HCU-associated isolates we analyzed, 112 (95.7%) were HCU1 cluster, 3 (2.6%) were HCU2 cluster, and 2 isolates (1.7%) were not in a major clade.
Whole-genome sequencing of US HCU-associated M. chimaera isolates and their comparisons with global HCU-associated isolates provides further evidence for point-source contamination and worldwide dissemination of a M. chimaera strain (3–5). Twenty-two of 24 (92%) US patient isolates associated with HCU exposure during cardiac surgery phylogenetically clustered with international HCU-derived and post–cardiac surgery patient isolates, including those from Australia, Europe, and New Zealand (HCU1). None of the 8 US non–HCU-associated isolates were genetically similar to the HCU1 or HCU2 clusters. Isolates from US post–cardiac surgery patients were genetically more similar to isolates derived from international LivaNova 3T HCUs (mean pairwise distance 4 SNPs) than M. chimaera isolates from US patients without a history of cardiac surgery (mean pairwise distance 511 SNPs). This evidence supports the hypothesis that US post–cardiac surgery M. chimaera infections were acquired from exposure to factory-contaminated HCUs rather than local populations of waterborne M. chimaera in each hospital.
Our analyses revealed that all US M. chimaera isolates associated with LivaNova 3T HCU exposure genetically cluster with HCU1 genotype isolates implicated in the global outbreak of post–cardiac surgery M. chimaera infections. The HCU2 cluster was not observed in the United States but included 2 isolates from HCUs in Australia, as well as a representative genotype of M. chimaera found in HCUs in Europe and at the HCU production site. Consistent with previous findings, this finding suggests the international circulation of a second, less plentiful, strain in the manufacturing site water system (8).
These observations support the hypothesis that the LivaNova 3T HCU design provided suitable conditions for both NTM colonization and aerosolization, particularly by M. chimaera. Even though production site contamination with M. chimaera has been confirmed, the medical community needs to remain alert for HCU-associated NTM infections involving other species (4). HCUs are vulnerable to contamination from in-hospital water sources, use of improper water sources, and improper maintenance, each of which may increase the risk of infection by NTM (including M. abscessus, M. chelonae, and M. gordonae, in addition to M. chimaera) (6). Contaminated HCUs may contain NTM-contaminated biofilms. Furthermore, water from the LivaNova 3T HCUs can become aerosolized during normal function, leading to introduction of potentially infectious particles into the sterile field, onto graft materials, or into the open chest cavity during cardiac surgery. The death rate for HCU-associated M. chimaera infections has been reported to be 50%; the latent period to diagnosis can be up to 5 years postsurgery (4,6,7,9,10), further emphasizing necessary diligence on the part of physicians and cardiac surgery patients to monitor for symptoms of disseminated NTM infection.
Our study has some limitations in methodology. We did not obtain samples from every US hospital that reported LivaNova 3T HCU–associated M. chimaera cases; no submitting hospital collected all 3 types of samples (HCUs, non-HCU samples, and suspected case-patients); and HCU samples were not collected by a single person or according to a standardized collection protocol. Despite these limitations, this analysis of US HCU-associated M. chimaera isolates clearly shows the clustering of isolates from epidemiologically linked US cases to international LivaNova 3T HCU M. chimaera isolates and the HCU1 genotype found within the LivaNova manufacturing site.
In conclusion, the application of WGS has advanced our understanding of M. chimaera present in US LivaNova 3T HCUs and patient cases after the initial analysis of suspected cases in Pennsylvania and Iowa. Given the innate drug resistance and the high death rate of HCU-associated M. chimaera infections, it remains imperative for hospitals to follow Food and Drug Administration guidelines (9) and the manufacturer’s instructions to minimize the risk of patient infection. In addition, clinicians should monitor patients who have had cardiac surgery using LivaNova 3T HCUs for signs and symptoms of NTM infection to enable early diagnosis and treatment.
Dr. Hasan is a researcher at the Center for Genes, Environment and Health at National Jewish Health, Denver, Colorado, USA. His main research focuses on nontuberculous Mycobacteria comparative genomics to discover genetic markers associated with pathoadaptation and conduct outbreak surveillance.
N.A.H., L.E.E., C.L.D., and M.S. acknowledge the support of the National Jewish Health NTM Center for Excellence and the Colorado Cystic Fibrosis Foundation Research Development Program (grant no. NICK15R0); M.S., N.A.H., and L.E.E. acknowledge the support of the Boettcher Foundation; and A.L., A.L.H., M.B.C., J.F.P., H.M.M., K.M.P., and K.A.P. acknowledge the support of the US Centers for Disease Control and Prevention.
- van Ingen J, Turenne CY, Tortoli E, Wallace RJ Jr, Brown-Elliott BA. A definition of the Mycobacterium avium complex for taxonomical and clinical purposes, a review. Int J Syst Evol Microbiol. 2018;68:3666–77.
- Tan N, Sampath R, Abu Saleh OM, Tweet MS, Jevremovic D, Alniemi S, et al. Disseminated Mycobacterium chimaera infection after cardiothoracic surgery. Open Forum Infect Dis. 2016;3:ofw131.
- Achermann Y, Rössle M, Hoffmann M, Deggim V, Kuster S, Zimmermann DR, et al. Prosthetic valve endocarditis and bloodstream infection due to Mycobacterium chimaera. J Clin Microbiol. 2013;51:1769–73.
- Sax H, Bloemberg G, Hasse B, Sommerstein R, Kohler P, Achermann Y, et al. Prolonged outbreak of Mycobacterium chimaera infection after open-chest heart surgery. Clin Infect Dis. 2015;61:67–75.
- Kohler P, Kuster SP, Bloemberg G, Schulthess B, Frank M, Tanner FC, et al. Healthcare-associated prosthetic heart valve, aortic vascular graft, and disseminated Mycobacterium chimaera infections subsequent to open heart surgery. Eur Heart J. 2015;36:2745–53.
- Food and Drug Administration. Nontuberculous mycobacterium (NTM) infections associated with heater-cooler devices (HCD) during cardiothoracic surgery. 2016 Meeting Materials of the Circulatory System Devices Panel [cited 2018 Mar 29]. https://www.fda.gov/AdvisoryCommittees/CommitteesMeetingMaterials/MedicalDevices/MedicalDevicesAdvisoryCommittee/CirculatorySystemDevicesPanel/ucm485091.htm
- WellSpan Health System. WellSpan York Hospital notifies open-heart surgery patients of possible infection risk. 2015 [cited 2018 Mar 30]. https://www.wellspan.org/news/story/15810
- Haller S, Höller C, Jacobshagen A, Hamouda O, Abu Sin M, Monnet D, et al. Contamination during production of heater-cooler units by Mycobacterium chimaera potential cause for invasive cardiovascular infections: results of an outbreak investigation in Germany, April 2015 to February 2016. Euro Surveill. 2016;21:30215.
- van Ingen J, Kohl TA, Kranzer K, Hasse B, Keller PM, Katarzyna Szafrańska A, et al. Global outbreak of severe Mycobacterium chimaera disease after cardiac surgery: a molecular epidemiological study. Lancet Infect Dis. 2017;17:1033–41.
- Hasan NA, Lawsin A, Perry KA, Alyanak E, Toney NC, Malecha A, et al. Complete genome sequence of Mycobacterium chimaera strain CDC 2015-22-71. Genome Announc. 2017;5:e00693-17.
- Perkins KM, Lawsin A, Hasan NA, Strong M, Halpin AL, Rodger RR, et al. Notes from the field: Mycobacterium chimaera contamination of heater-cooler devices used in cardiac surgery—United States. MMWR Morb Mortal Wkly Rep. 2016;65:1117–8.
- Williamson D, Howden B, Stinear T. Mycobacterium chimaera spread from heating and cooling units in heart surgery. N Engl J Med. 2017;376:600–2.
- Chand M, Lamagni T, Kranzer K, Hedge J, Moore G, Parks S, et al. Insidious risk of severe Mycobacterium chimaera infection in cardiac surgery patients. Clin Infect Dis. 2017;64:335–42.
TableCite This Article
- Page created: February 19, 2019
- Page last updated: February 19, 2019
- Page last reviewed: February 19, 2019
- Centers for Disease Control and Prevention,
National Center for Emerging and Zoonotic Infectious Diseases (NCEZID)
Office of the Director (OD)